    DATAFORTH CORPORATION                         Phone: (520) 741-1404
    3331 E. Hemisphere Loop                       Fax:   (520) 741-0762
    Tucson, AZ  85706  USA                        email: info@dataforth.com

                                TEST DATA SHEET
    ~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~
    Date:     05-15-2018
    Model:    SCM5B38-05
    SN:       129093-16

                                 ACCURACY TEST

                    Calculated       Measured
      Vin (mV)        Vout (V)        Vout (V)*      Error (%)       Status
    ==========      ==========      ==========      =========      ========
     -20.000         -5.000          -5.000          -0.005         PASS
      -9.999         -2.500          -2.499          +0.002         PASS
      -0.001         -0.000          +0.001          +0.016         PASS
     +10.001         +2.500          +2.502          +0.018         PASS
     +20.000         +5.000          +5.000          +0.003         PASS

                              FINAL TEST RESULTS

           Parameter         Measured Value       Specification      Status
    ======================== ===============  =====================  ======
    Supply Current, Nom              29.0 mA              < 75   mA  PASS
    Supply Current, Max             130.6 mA             < 194   mA  PASS
    Output Resistance                22 ohms            < 55   ohms  PASS
    Exc. Voltage                    10.001 V       10.0+/-0.003   V  PASS
    Exc. Load Reg.                  5 ppm/mA         +/-11   ppm/mA  PASS
    Vout Reg. w/ Load                  0.0 %             +/-0.1   %  PASS
    Exc. Current Limit               54.4 mA              < 63   mA  PASS
    Linearity                        0.010 %           +/- 0.03   %  PASS
    Accuracy                         0.018 %           +/- 0.08   %  PASS
    Supply Sensitivity              0.8 uV/%        +/-  2.4   uV/%  PASS
    Frequency Response               29.2 dB            27+/-7   dB  PASS
    Output Noise                   566 uVrms         < 4000   uVrms  PASS
    240 VAC Withstand                                                 PASS
    Hi-Pot                                                            PASS
    _______________________________________________________________________
                                  Check List

    Module Appearance:     __X__            Mounting Screw:          __X__

    Pins Straight:         __X__            Module Header:           __X__


    It is hereby certified that the above product is in conformance with
    all requirements to the extent specified.  This product is not
    authorized or warranted for use in life support devices and/or systems.

    * NIST traceable calibration certificates support Measured Value data.
      Calibration services are available through ANSI/NCSL Z540-1 and
      ISO Guide 25 Certified Metrology Labs.
